nasa cdf (version 3) {
// global attributes:
		:Project = "RBSP>Radiation Belt Storm Probes" ;
		:Source_name = "RBSPB>Radiation Belt Storm Probes B" ;
		:Discipline = "Solar Physics>Heliospheric Physics" ;
		:Discipline = "Space Physics>Magnetospheric Science" ;
		:Data_type = "L2>Level 2 Data" ;
		:Descriptor = "EMFISIS>Electric and Magnetic Field Instrument Suite and Integrated Science" ;
		:File_naming_convention = "source_datatype_descriptor_yyyyMMdd" ;
		:Data_version = "1.6.1" ;
		:PI_name = "Craig Kletzing" ;
		:PI_affiliation = "University of Iowa" ;
		:TEXT = "Three Axis Electric Field (EU, EV, EW) and Three Axis Magnetic field (BU, BV, BW)  continuous burst waveforms208896 samples @ 35 kS/sec...Calibration Notes:.Warning: All L2 Waveform products are calibrated in amplitude at 1kHz only.  There is no phase calibration applied at this stage...Before using these waveforms to process wave parameters, please follow the calibration process described below or the results will be wrong...Level-2 (L2) Waveform data is amplitude-calibrated at 1 kHz.  But there are amplitude deviations at other frequencies, and there are phase shifts which are not reflected in the L2 data at all. The phase shift is a frequency-dependent shift in the phase of the observed wave relative to the input wave, tantamount to a time delay at that frequency. . .The L2 data can be adjusted over frequency by applying dimensionless complex factors over frequency immediately after Fourier transforming the L2 data. The Variables BCalibrationCoef and ECalibration Coef in the cdf consists of a table for the B sensors and a table for the E sensors. Each table has 5600 X 2 entries. The  first column is the real component and the second is the imaginary component of the calibration, extending from 2.13623 to 11962.89 Hz, in steps of 2.13623 Hz  (Variable: CalFrequencies contains the associated frequencies of the calibration coeficients). Above the highest frequency the WFR filters roll off; no calibration measurements exist, but one could apply the last value to any frequencies above that.. .The table is constructed assuming 16384 data points are to be FFT'ed. If fewer than that are to be transformed, then the table can be decimated to accommodate a shorter data set. The procedure is;  FFT the L2 data at the desired resolution and then perform a complex multiplication of the FFT'ed dataset  and the E or B adjustment table.. .If comparing results of the frequency-adjusted L2 data with the onboard survey spectra, note that the L2 data has units of volts/meter and nanoTesla for E and B respectively, whereas the onboard survey spectra have units of RMS volts/meter and RMS nanoTesla.." ;
		:Instrument_type = "Radio and Plasma Waves (space)" ;
		:Instrument_type = "Electric Fields (space)" ;
		:Instrument_type = "Magnetic Fields (space)" ;
		:Mission_group = "RBSP" ;
		:Logical_source = "rbspb_emfisis_l2" ;
		:Logical_file_id = "rbsp-b_WFR-waveform-continuous-burst_emfisis-L2_20180613T22_v1.6.1.cdf" ;
		:Logical_source_description = "WFR Spectral Matrix" ;
		:Time_resolution = "6 seconds" ;
		:Rules_of_use = "From the Data Use Policy: http://emfisis.physics.uiowa.edu/data/use_policy Invite to be on the authorship: *The Principal Investigator, or his/her designate, of the team that generated the data *Any other contributors and scientists (from your team or other tems) who significantly contributed to the specific work described in your publication Allow any members of the author list to request to have their name removed from the publication authorship list as they so choose. Notify the PI of the team that produced the data that you are working on a paper and the topic of the paper well before circulating a first draft. Circulate a copy to all co-authors at least two weeks before you intend to submit and be willing to wait on submission if there are concerns raised by a member of the instrument team." ;
		:Generated_by = "University of Iowa" ;
		:Generation_date = "2018-06-14" ;
		:SPICE_kernels_used = "naif0012.tls rbspa_sclk_2088.tsc rbspb_sclk_2088.tsc em2_epoch_1980.tsc" ;
		:Input_files_used = "/var/EMFISIS-SOC/OUT/fm2/L1/2018/06/13/rbsp-b_WFR-waveform-continuous-burst_emfisis-L1_20180613T22_v1.6.1.cdf" ;
		:Software_used = "wfr_waveform_calibrator.pro, version1.2.1" ;
		:LINK_TEXT = "EMFISIS data" ;
		:LINK_TITLE = "EMFISIS" ;
		:HTTP_LINK = "http://emfisis.physics.uiowa.edu/" ;
		:RBSP_APID = "0x2b0" ;
		:RBSP_dataType = "WFRwaveform" ;
// variable attributes, one line per entry (which variable each belongs to is not shown):
		CATDESC = "Default time" ;
		CATDESC = "BuSamples" ;
		CATDESC = "BvSamples" ;
		CATDESC = "BwSamples" ;
		CATDESC = "EuSamples" ;
		CATDESC = "EvSamples" ;
		CATDESC = "EwSamples" ;
		CATDESC = "LWEzGainW" ;
		CATDESC = "LWExEyGainUV" ;
		CATDESC = "SCMGain" ;
		CATDESC = "Sun Pulse Time Tag" ;
		CATDESC = "timeOffsets" ;
		CATDESC = "size of FFT" ;
		CATDESC = "Bu Power Spectra" ;
		CATDESC = "Bv Power Spectra" ;
		CATDESC = "Bw Power Spectra" ;
		CATDESC = "Eu Power Spectra" ;
		CATDESC = "Ev Power Spectra" ;
		CATDESC = "Ew Power Spectra" ;
		CATDESC = "MET" ;
		CATDESC = "APID" ;
		CATDESC = "B Calibration Coefficients" ;
		CATDESC = "E Calibration Coefficients" ;
		CATDESC = "CalibrationCoef_LABL" ;
		CATDESC = "Calibration Frequencies" ;
		CATDESC = "Wave Power Quality Flag" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_0 = "CalFrequencies" ;
		DEPEND_0 = "CalFrequencies" ;
		DEPEND_0 = "Epoch" ;
		DEPEND_1 = "timeOffsets" ;
		DEPEND_1 = "timeOffsets" ;
		DEPEND_1 = "timeOffsets" ;
		DEPEND_1 = "timeOffsets" ;
		DEPEND_1 = "timeOffsets" ;
		DEPEND_1 = "timeOffsets" ;
		DEPEND_1 = "CalibrationCoef_LABL" ;
		DEPEND_1 = "CalibrationCoef_LABL" ;
		DICT_KEY = "" ;
		DICT_KEY = "flag" ;
		DICT_KEY = "flag" ;
		DICT_KEY = "flag" ;
		DICT_KEY = "flag" ;
		DICT_KEY = "flag" ;
		DICT_KEY = "flag" ;
		DISPLAY_TYPE = "waveform" ;
		DISPLAY_TYPE = "waveform" ;
		DISPLAY_TYPE = "waveform" ;
		DISPLAY_TYPE = "waveform" ;
		DISPLAY_TYPE = "waveform" ;
		DISPLAY_TYPE = "waveform" ;
		DISPLAY_TYPE = "spectrogram" ;
		DISPLAY_TYPE = "spectrogram" ;
		DISPLAY_TYPE = "spectrogram" ;
		DISPLAY_TYPE = "spectrogram" ;
		DISPLAY_TYPE = "spectrogram" ;
		DISPLAY_TYPE = "spectrogram" ;
		FIELDNAM = "Epoch" ;
		FIELDNAM = "BuSamples" ;
		FIELDNAM = "BvSamples" ;
		FIELDNAM = "BwSamples" ;
		FIELDNAM = "EuSamples" ;
		FIELDNAM = "EvSamples" ;
		FIELDNAM = "EwSamples" ;
		FIELDNAM = "LWEzGainW" ;
		FIELDNAM = "LWExEyGainUV" ;
		FIELDNAM = "SCMGain" ;
		FIELDNAM = "Epoch" ;
		FIELDNAM = "timeOffsets" ;
		FIELDNAM = "fftSize" ;
		FIELDNAM = "BuSpec" ;
		FIELDNAM = "BvSpec" ;
		FIELDNAM = "BwSpec" ;
		FIELDNAM = "EuSpec" ;
		FIELDNAM = "EvSpec" ;
		FIELDNAM = "EwSpec" ;
		FIELDNAM = "MET" ;
		FIELDNAM = "APID" ;
		FIELDNAM = "BCalibrationCoef" ;
		FIELDNAM = "ECalibrationCoef" ;
		FIELDNAM = "CalibrationCoef_LABL" ;
		FIELDNAM = "CalFrequencies" ;
		FIELDNAM = "chorusQF" ;
		FILLVAL = (24 numeric entries, not shown) ;
		FORMAT = "E12.2" ;
		FORMAT = "E12.2" ;
		FORMAT = "E12.2" ;
		FORMAT = "E12.2" ;
		FORMAT = "E12.2" ;
		FORMAT = "E12.2" ;
		FORMAT = "I3" ;
		FORMAT = "I3" ;
		FORMAT = "I3" ;
		FORMAT = "E12.2" ;
		FORMAT = "I8" ;
		FORMAT = "E12.4" ;
		FORMAT = "E12.4" ;
		FORMAT = "E12.4" ;
		FORMAT = "E12.4" ;
		FORMAT = "E12.4" ;
		FORMAT = "E12.4" ;
		FORMAT = "E12.2" ;
		FORMAT = "I5" ;
		FORMAT = "E12.2" ;
		FORMAT = "E12.2" ;
		FORMAT = "A13" ;
		FORMAT = "E12.2" ;
		FORMAT = "I3" ;
		LABLAXIS = "Time" ;
		LABLAXIS = "Time" ;
		LABLAXIS = "interval time" ;
		LABLAXIS = "power" ;
		LABLAXIS = "power" ;
		LABLAXIS = "power" ;
		LABLAXIS = "power" ;
		LABLAXIS = "power" ;
		LABLAXIS = "power" ;
		LABLAXIS = "APID" ;
		LABLAXIS = "B Calibration Coefficients" ;
		LABLAXIS = "E Calibration Coefficients" ;
		LABLAXIS = "CalibrationCoef_LABL" ;
		LABLAXIS = "Calibration Frequencies" ;
		UNITS = "ns" ;
		UNITS = "nT" ;
		UNITS = "nT" ;
		UNITS = "nT" ;
		UNITS = "V/m" ;
		UNITS = "V/m" ;
		UNITS = "V/m" ;
		UNITS = "ns" ;
		UNITS = "ns" ;
		UNITS = "" ;
		UNITS = "nT^2/Hz" ;
		UNITS = "nT^2/Hz" ;
		UNITS = "nT^2/Hz" ;
		UNITS = "V^2/m^2/Hz" ;
		UNITS = "V^2/m^2/Hz" ;
		UNITS = "V^2/m^2/Hz" ;
		UNITS = "Hz" ;
		VALIDMIN = (24 numeric entries, not shown) ;
		VALIDMAX = (24 numeric entries, not shown) ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		VAR_TYPE = "support_data" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		SCALETYP = "linear" ;
		VAR_NOTES = "Attenuator select indicator for W axis.  0 - Off, 1 - On.   Switches in a 19 dB attenuator.   L2 and greater data has the attenuator gains folded into data to provide correct science units." ;
		VAR_NOTES = "Attenuator select indicator for U and V axis.  0 - Off, 1 - On.   Switches in a 19 dB attenuator.  L2 and greater data has the attenuator gains folded into data to provide correct science units." ;
		VAR_NOTES = "Attenuator select indicator for MSC.  0 - Off, 1 - On.   Switches in a 19 dB attenuator.  L2 and greater data has the attenuator gains folded into data to provide correct science units." ;
		VAR_NOTES = "" ;
		VAR_NOTES = "" ;
		VAR_NOTES = "" ;
		VAR_NOTES = "" ;
		VAR_NOTES = "" ;
		VAR_NOTES = "" ;
		VAR_NOTES = "" ;
		VAR_NOTES = "Calibration Coeficients for all three B axes waveforms.   Coefficients are real and imagniary components that should be applied in frequency space.   Reference CalFrequency variable for center band frequency.   First column is the real component and the second is the imaginary component." ;
		VAR_NOTES = "Calibration Coefficients for all three E axes waveforms.   Coefficients are real and imagniary components that should be applied in frequency space.   Reference CalFrequency variable for center band frequency.   First column is the real component and the second is the imaginary component." ;
		VAR_NOTES = "Calibration Frequencies .To properly correct for phase and amplitude, the calibrations should be applied in frequency space." ;
		VAR_NOTES = "Onboard calculated Quality Factor A quality factor is built using one or more of the onboard Waves space weather parameters to indicate activity. The space weather parameters reflect the intensity in three spectral bands in both the electric and magnetic channels of the Waves instruments. These correspond to, approximately, lower band chorus, upper band chorus (these separated by one-half of the electron cyclotron frequency) and a low-frequency band below the lower hybrid frequency." ;
		MONOTON = "INCREASE" ;
		MONOTON = "INCREASE" ;
		VIRTUAL = "true" ;
		VIRTUAL = "true" ;
		VIRTUAL = "true" ;
		VIRTUAL = "true" ;
		VIRTUAL = "true" ;
		VIRTUAL = "true" ;
		FUNCT = "fftPower" ;
		FUNCT = "fftPower" ;
		FUNCT = "fftPower" ;
		FUNCT = "fftPower" ;
		FUNCT = "fftPower" ;
		FUNCT = "fftPower" ;
		COMPONENT_0 = "BuSamples" ;
		COMPONENT_0 = "BvSamples" ;
		COMPONENT_0 = "BwSamples" ;
		COMPONENT_0 = "EuSamples" ;
		COMPONENT_0 = "EvSamples" ;
		COMPONENT_0 = "EwSamples" ;
		COMPONENT_1 = "fftSize" ;
		COMPONENT_1 = "fftSize" ;
		COMPONENT_1 = "fftSize" ;
		COMPONENT_1 = "fftSize" ;
		COMPONENT_1 = "fftSize" ;
		COMPONENT_1 = "fftSize" ;
		AVG_TYPE = "" ;
		AVG_TYPE = "" ;
		AVG_TYPE = "" ;
		AVG_TYPE = "" ;
		AVG_TYPE = "" ;
		AVG_TYPE = "" ;
}